Excessive Anteversion Leads to Failure at 3 Years Due to Impingement as Evidenced by Twin Notches in Ti6A4V Stem
Keywords:S-ROM, metal-on-metal, impingement, notches, total hip
AbstractA 63-year old female with bilateral hip replacements was referred to our clinic for pain and elevated metal ions. Her left hip had been revised earlier. The right hip had an SROM Ti6Al4V stem implanted with a 28mm head, a 28mm CoCr liner and Pinnacle Ti6Al4V shell. The patient reported pain, numbness, tingling, and repeated clicking and popping sensations with gait. She specifically noted that her hip would freeze while walking and could pop rising from a chair. Repeated metal ion levels showed Co (blood)17ppb, Cr (serum) 21ppb, and Ti (blood) at 69ppb. CT-images of right hip revealed femoral stem anteversion was 43° and cup anteversion was 40°, for a combined anteversion of 83°. The right hip was revised 3.5 years postoperatively for persistent pain and elevated metal ions. At surgery, large twin notches were evident on her posterior femoral neck and 10mm-wide scalloped damage was evident in the rim of the Ti6A4V shell. SEM-imaging revealed contaminating layers on CoCr head containing elements Al, V and Ti. These indicated that titanium-alloy particles liberated by cup-to-neck impingements had transferred to the CoCr bearings. Our intent in this case was not to document that a MOM bearing produced impingement damage, because this case clearly implicated adverse surgical positioning. Rather, the intent was to document sequelae likely in a THA case that has a metal cup impinging on a metal femoral neck. In particular, twin notches on the femoral neck indicated that this patient was routinely impinging her Ti6Al4V shell against the Ti6Al4V neck and also subluxing her femoral head out of the cup. These signs are a clear indication that one or both components must be revised, as opposed to simply replacing the CoCr liner with a revision polyethylene liner.
R.L. Barrack, T.P. Schmalzried, Impingement and rim wear associated with early osteolysis after a total hip replacement : a case report, J Bone Joint Surg Am, 84-A (2002) 1218-1220.
D. Gambera, S. Carta, E. Crainz, M. Fortina, P. Maniscalco, P. Ferrata, Metallosis due to impingement between the socket and the femoral head in a total hip prosthesis. A case report, Acta Biomed Ateneo Parmense, 73 (2002) 85-91.
M. Yamaguchi, T. Akisue, T.W. Bauer, Y. Hashimoto, The spatial location of impingement in total hip arthroplasty, J Arthroplasty, 15 (2000) 305-313.
M. Earll, T. Fehring, W.L. Griffin, J.B. Mason, T.H. McCoy, Early osteolysis associated with trunion-liner impingement, Clin Orthop, (2004) 153-156.
K. Onda, S. Nagoya, M. Kaya, T. Yamashita, Cup-neck impingement due to the malposition of the implant as a possible mechanism for metallosis in metal-on-metal total hip arthroplasty, Orthopedics, 31 (2008) 396.
M. Manaka, I.C. Clarke, A. Gustafson, A. Imakiire, Impingement in Ceramic Hip Patient: A retrieval and Wear-Scar Analysis of Femoral Head., Key Eng. Mater, 240-242 (2003) 847-848.
F. Randelli, L. Banci, A. D’Anna, O. Visentin, G. Randelli, Cementless Metasul metal-on-metal total hip arthroplasties at 13 years, Journal of Arthroplasty, 27 (2012) 186 - 192.
M. Kligman, B.D. Furman, D.E. Padgett, T.M. Wright, Impingement contributes to backside wear and screw-metallic shell fretting in modular acetabular cups, J Arthroplasty, 22 (2007) 258-264.
W.Y. Shon, T. Baldini, M.G. Peterson, T.M. Wright, E.A. Salvati, Impingement in total hip arthroplasty a study of retrieved acetabular components, J Arthroplasty, 20 (2005) 427-435.
R.M. Hall, P. Siney, A. Unsworth, B.M. Wroblewski, Prevalence of impingement in explanted Charnley acetabular components, J Orthop Sci, 3 (1998) 204-208.
C.I. Esposito, W.L. Walter, A. Roques, M.A. Tuke, B.A. Zicat, W.R. Walsh, W.K. Walter, Wear in alumina-on-alumina ceramic bearings. A retrieval analysis of edge loading, J Bone Joint Surg Am, 7 (2012) 901-907.
I.C. Clarke, T.K. Donaldson, M.D. Burgett, E.J. Smith, J. Bowsher, C. Savisaar, A. John, J.Y. Lazennec, E. McPherson, C.L. Peters, Normal and Adverse Wear Patterns Created
In-vivo on MOM Surfaces - a retrieval study representing four vendors, in: S.M. Kurtz, S.A. Greenwald, W.M. Mihalko, J.A. Lemons (Eds.) ASTM Symposium on Metal-on-Metal Total Hip Replacement Devices, ASTM International, West Conshohocken, PA, 2013 pp. 157-192.
E.J. McPherson, I.C. Clarke, T.K. Donaldson, M.D. Burgett, A MOM dislocator case demonstrating importance of impingement damage in triggering adverse 3rd-body abrasive wear risks, in: Orthopedics Today, Jan 13 - 16, Hawaii, 2013.
E.J. McPherson, I.C. Clarke, T.K. Donaldson, Lesson learned from retrieval analysis of a dislocating, large diameter MoM revision THA - A case report, Reconstructive Review, 2 (2012) 10 -14.
E. McPherson, C. Peters, Correction of Femoral Acetabular Impingement at the Time of Primary Large Diameter Metal-Metal THA, in: AAOS New Orleans, LA, 2010.
K. Bjorgul, W.N. Novicoff, S.T. Andersen, O.R. Ahlund, A. Bunes, M. Wiig, K. Brevig, High rate of revision and a high incidence of radiolucent lines around Metasul metal-on-metal total hip replacements, Bone Joint J, 95-B (2013).
V. Eswaramoorthy, P. Moonot, Y. Kalairajah, L.C. Biant, R.E. Field, The Metasul metal-on-metal articulation in primary total hip replacement: clinical and radiological results at ten years, J Bone Joint Surg Br, 90 (2008) 1278-1283.
J.Y. Lazennec, P. Boyer, J. Poupon, M.A. Rousseau, C. Roy, P. Ravaud, Y. Catonne, Outcome and serum ion determination up to 11 years after implantation of a cemented metal-on-metal hip prosthesis, Acta Orthop, 80 (2009) 168-173.
H. Iida, E. Kaneda, H. Takada, K. Uchida, K. Kawanabe, T. Nakamura, Metallosis due to impingement between the socket and the femoral neck in a metal-on-metal bearing total hip prosthesis. A case report, J Bone Joint Surg Am, 81 (1999) 400-403.
McMurtrie, R. Abhijit, A.R. Guha, J.R. Wootton, Loose Metasul liner causing partial amputation of the neck of the femoral component, Journal of Arthroplasty, 24 (2009) 159.e151 - 153.
Malik, L.D. Dorr, W.T. Long, Impingement as a mechanism of dissociation of a metasul metal-on-metal liner, J Arthroplasty, 24 (2009) 323 e313-326.
T.H. Eickmann, I.C. Clarke, A. Gustafson, Squeaking in a ceramic on ceramic total hip, in: H. Zippel, M. Dietrich (Eds.) Bioceramics in Joint Arthroplasty, Steinkopff Verlag, Darmstadt, Berlin, Germany, 2003, pp. 187 - 192.
Clarke I C, Lazennec J Y, Brusson A, Savisaar C, Bowsher J G, Burgett M, et al. Risk of Impingement and Third-body Abrasion With 28-mm Metal-on-metal Bearings. Clin Orthop Relat Res 2014; 472 (2): 497-508.
Halim T, Burgett M, Donaldson TK, et al. Profiling the third-body wear damage produced in CoCr surfaces by bone cement, CoCr, and Ti6Al4V debris: a 10-cycle metal-on-metal simulator test. Proc Inst Mech Eng H 2014;228:703–713.
Halim T, Clarke I C, Burgett-Moreno M, Donaldson T, Savisaar C, Bowsher J. A simulator study of adverse wear with metal and cement debris contamination in metal-on-metal hip bearings. Bone Joint Res 2015;4:29-37.
How to Cite
Copyright (c) 2015 Thomas Donaldson, Michelle Burgett-Moreno, Ian Clarke
This work is licensed under a Creative Commons Attribution-NonCommercial 4.0 International License.
Copyright and License Agreement:
Authors who publish with the Reconstructive Review agree to the following terms:
- Authors retain copyright and grant the journal right of first publication with the work. Reconstructive Review follows the Creative Commons Attribution-NonCommercial CC BY-NC. This license allows anyone to download works, build upon the material, and share them with others for non-commercial purposes as long as they credit the senior author, Reconstructive Review, and the Joint Implant Surgery & Research Foundation (JISRF). An example credit would be: "Courtesy of (senior author's name), Reconstructive Review, JISRF, Chagrin Falls, Ohio". While works can be downloaded and shared they cannot be used commercially.
- Authors are able to enter into separate, additional contractual arrangements for the non-exclusive distribution of the journal's published version of the work (e.g., post it to an institutional repository or publish it in a book), with an acknowledgement of its initial publication in this journal.
- Authors are permitted and encouraged to post their work online (e.g., in institutional repositories or on their website) prior to and during the submission process, as it can lead to productive exchanges, as well as earlier and greater citation of published work.